Loading option module firmware using SFTP
You must have:
- Your meter’s option module firmware upgrade (.S19) file(s). Go to www.schneider-electric.com to download upgrade files.
The option module connected to your meter, and have access to your meter’s internal SFTP site.
Your meter’s digital and analog outputs may change state during a firmware upgrade. Option modules are offline during an option module firmware upgrade; they will not communicate to the meter and the outputs may change state during the upgrade process.
Do not use this device for critical control or protection applications where human or equipment safety relies on the operation of the control circuit.
Failure to follow these instructions can result in death, serious injury, or equipment damage.
- Connect to your meter’s internal SFTP site. Open the option module upgrade (optionModuleUpg) folder.
- Drag the option module firmware file into the option module
upgrade folder.The file is immediately transferred to the option module, and is not stored on the meter.
- Review the option module firmware upgrade details stored in the meter’s event log to confirm that the upload was successful.