Configuring status inputs using ION Setup
- Start ION Setup.
- Connect to your meter.
- Navigate to .
- Select a status input to configure and click Edit.
The setup screen for that status input is displayed.
- Enter a descriptive name for the status input’s Label.
- Configure the other setup parameters as required.
- Click Send to save your changes.
Status input setup parameters available through ION Setup
Parameter Values Description
Use this field to change the default label and assign a descriptive name to this status input.
Normal, Demand Sync, Input Metering
This field displays how the status input functions.
Normal: the status input is not associated with another meter function. The meter counts and records the number of incoming pulses normally.
Demand Sync: the status input is associated with one of the input sync demand functions. The meter uses the incoming pulse to synchronize its demand period with the external source.
Input Metering: the status input is associated with one of the input metering channels. The meter counts and records the number of incoming pulses and related consumption data associated with the pulses.
0 to 9999
Debounce is the time delay that compensates for mechanical contact bounce. Use this field to set how long (in milliseconds) the external signal must remain in a certain state to be considered a valid state change.
This field displays additional information if the status input is already associated with another meter function.