Configuring waveform capture using the display
- Navigate to .
- Enter the setup passcode (default is “0”), then press OK.
- Navigate to WFC.
- Move the cursor to point to the parameter you want to modify, then press Edit.
- Modify the parameters as required, then press OK.
- Press the up arrow to exit. Press Yes to save your changes.
Waveform capture setup parameters available using the display
Enables or disables the waveform capture on your meter
Samples Per Cycle
Select the samples per cycle based on the requirement
The pre cycles can be configured based on the samples per cycle
Maximum 1 – 8 pre cycles for 128 samples per cycle
Maximum 1 – 16 pre cycles for 64 samples per cycle
Maximum 1 – 32 pre cycles for 32 samples per cycle
Maximum 1 – 64 pre cycles for 16 samples per cycle
Select the pre cycles based on the requirement
The Post-cycles will be automatically configured based on the Pre-cyclesNOTE: It is recommended for the end user to configure equal number of pre and post cycles to visualize sag/swell events in waveform.
Pre-cycles: Number of cycles that will be captured in waveforms prior to waveform trigger
Post-cycles : Number of cycles that will be captured in waveforms post waveform trigger